Download FLUKE PM PM PM PM 2X2MV,MHZ OSCILLOSOPE SM service manual & repair info for electronics experts. Re: Fluke PM manual needed (MHz scope) The service manual for Fluke / Philips PM PM PM PM can be found. PM ). Deflection Manual positioning: Horizontal, Vertical, Both . Web access: com.
|Country:||Central African Republic|
|Published (Last):||10 August 2015|
|PDF File Size:||3.86 Mb|
|ePub File Size:||10.77 Mb|
|Price:||Free* [*Free Regsitration Required]|
Input Characteristics Note 1: Send a copy to my email address. However, many of the standard electronic components can be obtained from other local manuxl. Page Verify that the deviation from the ideal straight line does not exceed 0. Some of these voltages remain some time after disconnecting the instrument from the mains. Page – Maximum phase shift between horizontal a Please also ensure that: Use a 50Q termination.
Oscilloscope Fluke PM powersupply problem?
Page 3 – Use terminations. We are making things easier for you, auction specific shipping information coming soon.
Here, the accuracy of this range is checked. Verify that the delay difference between the two displayed signals is less than 0. Where satisfactory evidence is not provided, goods will be subject to UK VAT at the standard rate of export and the deposit will not be refunded. The other IC D is the peak- peak detector for the trigger level. In this test, the values of frequency and voltage are checked.
The smps is an ordinary flyback supply.
Fluke PM Analog Oscilloscope | eBay
Page – Probe Adjust signal; frequency and outpu Really don’t know why the mains fuse went. Any concluded contract of sale is made directly between the Vendor and the Buyer.
Sorry, bidding has ended on this item. Page Page – Horizontal deflection; X deflection coef To print the manual completely, please, download it. In the signal name list you find the itemnumber of the component that is source or destination.
Page Probe Adjust signal is distortion-free and has equal intensity. Table Of Contents 6. Also for this the oscilloscope must be sent to manuao nearest Fluke Service Center. The digital multimeter and oscilloscope are not required for this test.
Printed circuit board lay-out of external trigger unit Page The linearity of the vertical amplifier is checked by moving a signal with a fixed amplitude vertically over the entire screen area.
For the avoidance of doubt removal includes where applicable disconnection from the mains electricity supply and the detachment manuak any lots which are fixed. Input attenuators and calibrator. I was given an oscilloscope, a Fluke PM from a retired industrial electronics engineer. Over and under-voltage protection To protect the oscilloscope circuitry against over-voltage and the supply circuits against overload, the power supply is provided with a protection circuit.
Repeat the above settings and procedure for CH3 and CH4. This diagram shows the RC-branches and regulators N that give the supply voltages for the various circuits. Refer to section 8.
Fluke PM3082 Service Manual
Before turning on the instrument, ensure that it has been installed in accordance with the Installation Instructions, outlined in Section 2 of the Operation Guide. Check that the leads of the replacement part are clean and pre-tinned on the soldering places.
Venue Address Lefevre Way. This delay is verified in the following test. The IEEE connector is located at the rear panel of the oscilloscope.
The value of this shift is measured here. Page Verify that the jitter of the DTB is not more than 0.
Lambert Smith Hampton reserve the right to stop clearance if in their opinion, they believe it is being carried out in an unsafe manner or without the Method Statement and Risk Assessment. The input signal is a current and is applied to V and V