High speed – up to 15,rpm/21, x g * Whisper quiet brushless induction drive * Precise, microprocessor control * Automatic g-force conversion program. Hermle ZM-2 Microcentrifuge from Labnet International, Inc. A high performance microlitre centrifuge, offering a fullrange of features:digital display of time, rpm and RCFtwo acceleration and deceleration.
|Published (Last):||13 August 2014|
|PDF File Size:||4.3 Mb|
|ePub File Size:||13.84 Mb|
|Price:||Free* [*Free Regsitration Required]|
The temperature rise depends on the rotor swing-out or angle rotor ,bucket type, ambient temperature, running time and the speed of the rotor. If this is not done, they could scratch the protective coating of the rotor.
Hermle z m-2 Manuals
By pressing ‘print’ button you will a233 only current page. Figure 8 If the green “fast” control lamp is lit, the unit accelerates and decelerates fast. The digital indications on the display will light up when the power is on.
Open and reclose the lid, restart the centrifuge Error no.: The green control lamps 3 show which mode is activated. Set Or Change Rotor-code -Load the rotor symmetrically by inserting tubes into spaces opposite from each other.
Before operation, secure hsrmle rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Page of 20 Go.
Description – The rotor is not installed securely on the motor shaft. Width 28 cm Depth 37 cm Height If the green “slow”control lamp is lit, the unit accelerates and decelerates very softly below 2 rpm. Should this happen, the centrifuge, rotor, buckets, adapters and the rotor chamber must be thoroughly cleaned and all broken particles removed immediately.
Page 19 Error no.: Never stay in the safety zone of 30 cm around the centrifuge or deposit z323 materials inside this zone during centrifugation.
Don’t show me this message again. Got it, continue to print. Page 15 – Temperture Page 16 – Service and inspection m–2 centrifuge Page 17 – Cleaning of centrifuge after breakage of Overcurrent or undervoltage due to power supply fluctuations Action Page 14 – Keyboard – Starting the centrifuge – “qu Don’t have an account?
Centrifuges Z 233 M-2 -Hermle- Germany
The rotor must always be balanced in this fashion prior to operation. Mm-2 And Inspection Of Centrifuge 7. Commercially available tubes or bottles made of glass or plastic. Enter text from picture: Operation Operation Operation Operation Operation 4.
The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations. Keyboard – Starting The Centrifuge – “quick”-key 4.
Your Trusted Source for Used Lab Equipment
To open the centrifuge lid. Use only original spare parts! Power failure while rotor is in motion Action: To print the manual completely, please, download it. With the centrifuge lid open you can preset the operating time with the knob 3.
G Hermle Z M-2 Labnet Centrifuge W/rotor | eBay
General Information The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations. Table of Hermel Add to my manuals Add. Page 20 Error no.: On the digital display “time min-sec” the letters “ro” rotor appears. Preselection Of Operating Time 4.
There is a distinction between two different kinds of errors. During a run and with the centrifuge lid closed you have to press the “preset” key 1hold it and change operating z2333 by turning the knob 3. Comments to this Manuals Your Name. Turn the speed knob 1see figure 4 till the required rotor code appears on the d i s p l a y.